Yes absolutely, it can basically measure anything that flows (water, chemicals, slurries, mud etc.)
Yes, The Meter can measure gas and gas with up to at least 50 weight percentage liquid.
No, this is a flow Meter not a sand detector.
No, today it can only measure single and two phase flow.
No, if your wells are gas wells (with/without) small amounts of liquid we would like to have input on Pressure and temperature.
No, the Meter is designed specially for each pipe sizes.
If the Meter shall measure watercut, temperature is an important input. If process temperature is not available, such can be arrived from the internal temperature sensors mounted on the surface of the pipe inside the Meter.
We would like to have a minimum of 20 bar pressure where we install the Meters (Xsens can lower this by doing some extended calibrations).
1,5” to 30” (higher might be possible, but Xsens have to evaluate).
For pressure there is no limits. Temperature is 100 deg C at surface of pipe today.
Range is 0-100% watercut, accuracy within +/-1% abs.
If only flow rate is needed, we can measure such without any input data.
If water cut is to be measured in addition to flow rate, we need to measure Speed of Sound (SoS) of a sample of pure oil and water plus temperature and salinity of water is required. Xsens have handheld SoS measurement cells available so we can do this one-time SoS measurement in the field.